Required: Powder-free gloves and a spraychip® (P/N 413711SP).
1. Press Main Menu in the right-hand panel.
2. Press System. The Select User dialog opens.
3. Select your user name and press OK.
4. If your account is password-protected enter your password and press OK.
5. Select the Maintenance tab.
6. Press Replace spraychip® in the Mass Detector field and follow the instructions.
Update on Biotage’s COVID-19 Response Plan
Smarter Sample Preparation for Environmental Challenges
How does Biotage Act in the Corona Virus Situation?
Sample Preparation Techniques for Synthetic Benzodiazepines
Good Laboratory Practices Software Upgrade for Biotage® Extrahera™