"I synthesized some 18A, an 18-amino acid amphipathic peptide ... The HPLC purity analysis indicates that the peptide is now >95% pure. Pretty good if you ask me! ... I then connected two identical cartridges inline through the luer fittings. I used the exact same gradient, as measured in column volumes, the same flow rate (no change in linear velocity), and injected the same crude peptide volume for the new “double” cartridge purification and selected run."
Find out what she learned in the post Can I improve my peptide purification by increasing the column length? on The Peptide Synthesis Blog ».
Ten Years with Environmental SPE Workflow - a User's Perspective
How St. John's University Accelerated their Cancer Research
Application Video: Extraction of Polyfluorinated Compounds in Water
Yes, You Can Add Biotage® Sfär Columns to Your Isolera™ System
Two New Applications Notes - Sample Preparation of a Drugs of Abuse Panel in Human Hair