Flash chromatography can be a challenging technique for peptide purification due to the lower resolution achieved with large particles. While some may see this as a disadvantage, the significantly greater loading capacity gives me reason to make this work. So how can I achieve the high purity levels often accessed using traditional reversed-phase HPLC methods?
In this post, I’ll discuss using a step gradient for peptide purification. Step gradients are commonly used in normal-phase small molecule purification and typically improve the purification efficiency while reducing the overall purification time.
I have already discussed several strategies that may help improve the overall purification quality, including changing the mobile phase modifier and using mixed stationary phases, but one strategy often overlooked is the step gradient.
I started down this experimental path with a peptide that I struggled to purify with a linear gradient. Although some impurities were resolved, the analytical HPLC-MS showed a trailing shoulder, indicating an impurity was still present in my sample, Figure 1.
To design my step gradients, I used the Optimize button on my Biotage® Isolera™ Dalton 2000. This feature allows you to select a peak of interest collected during a linear gradient and designs a step gradient optimized for the selected compound of interest. You do have to be a little careful using this feature for peptides as the software will likely choose an initial condition which would not allow for good compound retention by the column. It does though provide a good reference point for which concentration of MeCN I should use to elute my peptide of interest.
As a first attempt, I naively thought that I could achieve my purification goals in a single step, jumping from my initial conditions of 30% up to 55% (the MeCN concentration recommended by the software optimization), Figure 2. Boy was I wrong. Stepping up to such a high MeCN concentration prevented the early eluting impurities from being resolved and separated out. As a result, everything co-eluted. On the bright side though, some later eluting impurities were removed and this gave me a starting point for how to further optimize the step gradient.
With each successive gradient change, I either added additional steps or changed the MeCN concentration held during a step, allowing impurities to more fully resolve before stepping to a higher MeCN concentration, Figure 3. The one thing I was diligent in keeping constant though was the length of each isocratic hold. I chose to hold each step for 3 column volumes to ensure that the cartridge was fully equilibrated at that condition such that any compounds could elute before another drastic change in MeCN.
While still not perfect, the last step gradient I tried showed marked improvement in the final purity of my peptide when compared to the linear gradient I started with, Figure 4. The most interesting observation was how much difference holding at 53% rather than 55% MeCN could improve the purity. It just goes to show that significant purity enhancements can be achieved for peptides with small changes in MeCN concentration.
These results clearly demonstrate the utility of using step gradients for peptide purification. Although the time savings often observed for small molecule purification were not retained here, the improvements in final peptide purity came without a significant increase in purification time.
Interested to learn more about how flash chromatography can reduce your purification time and maintain high purity? Follow the link below.